As an example. here is the Quick Start Guide for the M-DVRT-9: http://files.microstrain.com/Micro_Sub_DVRT_Quick_Start_Guide.pdf
Page 7 refers to the Slope and Offset that you will need to enter into the Smart Motherboard software.
In the Smart Motherboard software, click Tools.
Click Configuration and the Configuration screen appears.
Select the appropriate channel (remember that each channel, i.e., each DVRT and DEMOD –DVRT signal conditioner card in the Smart Motherboard is calibrated separately) by checking the Channel check box.
Select the Linear Radio Button.
Enter the Slope and Offset in the number scroll boxes.
Select None in the Peak Detect drop-down.
Change the Units from Volts to mm (for millimeters) by wiping through with your mouse.
Click Save As Default.
Click Return and you are ready to sample in millimeters.