Our MIP Monitor software, as written, will not allow you to save the 12 hard and soft iron calibration coefficients to a file, and then read that file back into the software to rewrite the values to the inertial sensor.
However, you cand accomplish the task by recording the values that are displayed (or take a screen grab).
This would be followed by using the data communications protocol EEPROM Write command to write the values back into the device.
This would require either a terminal program or the ability to write software, an understanding of how to construct the command packets, and a list of the EEPROM addresses involved.
Please contact your Lord MicroStrain support engineer for further details.